eFunda: Introduction to Diaphragm Pressure Gages
 Diaphragm Pressure Gages: Introduction
 Design Home Sensors Sensor Home Instruments/Devices Methods/Principles Displacement Stress & Strain Pressure U Tubes Bourdon Tubes Diaphragm Intro Diaphragm Theory McLeod Gages Intro McLeod Gages Theory Fluid Flow Flowmeter Temperature Resources Bibliography
 Home Membership Magazines Forum Search Member Calculators
 Materials Design Processes Units Formulas Math
 Overview The Diaphragm Pressure Gage uses the elastic deformation of a diaphragm (i.e. membrane) instead of a liquid level to measure the difference between an unknown pressure and a reference pressure. A typical Diaphragm pressure gage contains a capsule divided by a diaphragm, as shown in the schematic below. One side of the diaphragm is open to the external targeted pressure, PExt, and the other side is connected to a known pressure, PRef,. The pressure difference, PExt - PRef, mechanically deflects the diaphragm. Typical Diaphragm Pressure Gage The membrane deflection can be measured in any number of ways. For example, it can be detected via a mechanically-coupled indicating needle, an attached strain gage, a linear variable differential transformer (LVDT; see the schematic below), or with many other displacement/velocity sensors. Once known, the deflection can be converted to a pressure loading using plate theory. LVDT-Based Diaphragm Pressure Gage
Pros and Cons

 • Pros: - Much faster frequency response than U tubes. - Accuracy up to ±0.5% of full scale. - Good linearity when the deflection is no larger than the order of the diaphragm thickness. • Cons: - More expensive then other pressure sensors.

Glossary