Design Home Sensors Sensor Home Instruments/Devices Methods/Principles Displacement Stress & Strain Pressure Fluid Flow Flowmeter Temperature Thermocouple Intro Thermocouple Theory RTD Intro RTD Theory Thermistors Intro Thermistors Theory Pyrometers Intro Pyrometers Theory Resources Bibliography
Offshore

International news and technology for marine/offshore operations around the world.

Quality Magazine

Techniques to improve quality on the shop floor and in manufacturing planning.

Salary Expectation

8 things to know about the interview question "What's your salary expectation"?

Laser Focus World

Semiconductors, medical equipment, lasers, optics and aviation and aerospace.

more free magazines       Resistance-Temperature Relationship The resistance-temperature (R-T) relationship plays a central role in resistance temperature detectors (RTDs). The R-T relationship of some common RTD materials are illustratrated in the following schematic where the y-axis is the normalized resistance with respect to resistance at 0 °C (32 °F), x-axis is the temperature. Resistance-Temperature Relationshipfor some RTD materials For a given material, the resistance at any temperature can be obtained by curve-fitting the R-T curve. Suppose that a nth degree polynomial least-square curve-fit is used. The temperature function will then be, Some references suggest to express the resistance as a function of temperature (rather than temperature as a function of resistance), i.e., Although such a relationship is mathematically correct, its usefulness is limited. After all, we are interested in determining temperature from a resistance measurement.
 A Simplified Case Some materials have an almost linear R-T relationship within a certain temperature range, T1 < T < T2. Such a linear function would take the form, Rearranging to bring temperature out gives, where a is the average temperature coefficient of resistance in the (T1,T2) temperature range, i.e., a is the slope of the R-T line. Both the measured temperature and the reference temparature should be within the (T1,T2) temperature range, T1 < {T, TRef} < T2
Glossary