Design Home Sensors Sensor Home Instruments/Devices Methods/Principles Displacement LVDT Introduction LVDT Applications LVDT Theory RVDT Introduction Fotonic Introduction Fotonic Theory Eddy Current Intro Eddy Current Theory Stress & Strain Pressure Fluid Flow Flowmeter Temperature Resources Bibliography
NASA Tech Briefs

Innovations developed by NASA and its industry partners in a wide array of fields.

3D Scanners

A white paper to assist in the evaluation of 3D scanning hardware solutions.

Desktop Engineering

Design, simulation, test, prototyping and high performance computing.

Intro to Injection Molding

Provides a basic understanding of plastic injection molding presses, processes, and costs.

more free magazines
 Effective Depth An eddy current is a local electric current induced in a conductive material by the magnetic field produced by the active coil. This local electric current in turn induces a magnetic field opposite in sense to the one from the active coil and reduces the inductance in the coil. When the distance between the target and the probe changes, the impedance of the coil changes correspondingly. This change in impedance can be detected by a carefully arranged bridge circuit. The eddy currents are confined to shallow depths near the conductive target surface. Their effective depth is given by, where f is the excitation frequency of the circuit, µ is the magnetic permeability of the target material, and s is the conductivity of the target material. The target material must be at least three times thicker than the effective depth of the eddy currents to make the transducer successful. This is because the transducer assumes that the eddy currents are localized near the surface of a semi-infinite solid, and the actual eddy current amplitude decreases quadratically with distance.
 Effective Sensor Range In practice, the effective range of an eddy current transducer is given by the vendor suggested range offset from the target surface by 20%. For example, a 2.5 mm (0.1 in) range eddy current transducer is generally considered effective from 0.5 to 3 mm (0.2 to 1.2 in) from the target surface. The targeted flat surface area should not be smaller that the probe tip diameter. If the target surface is smaller than 50% of the probe diameter, output signals decrease substantially.