Design Home Sensors Sensor Home Instruments/Devices Methods/Principles Displacement Stress & Strain Pressure Fluid Flow Flowmeter Temperature Thermocouple Intro Thermocouple Theory RTD Intro RTD Theory Thermistors Intro Thermistors Theory Pyrometers Intro Pyrometers Theory Resources Bibliography
Autonomous Vehicle Engineering

The No. 1 media source for those developing the next generation mobility solutions.

Design-2-Part

How OEM's can make their parts better, faster, and more efficient.

Intro to Injection Molding

Provides a basic understanding of plastic injection molding presses, processes, and costs.

Desktop Engineering

Design, simulation, test, prototyping and high performance computing.

more free magazines       Resistance-Temperature Relationship The resistance-temperature (R-T) relationship plays a central role in resistance temperature detectors (RTDs). The R-T relationship of some common RTD materials are illustratrated in the following schematic where the y-axis is the normalized resistance with respect to resistance at 0 °C (32 °F), x-axis is the temperature. Resistance-Temperature Relationshipfor some RTD materials For a given material, the resistance at any temperature can be obtained by curve-fitting the R-T curve. Suppose that a nth degree polynomial least-square curve-fit is used. The temperature function will then be, Some references suggest to express the resistance as a function of temperature (rather than temperature as a function of resistance), i.e., Although such a relationship is mathematically correct, its usefulness is limited. After all, we are interested in determining temperature from a resistance measurement.
 A Simplified Case Some materials have an almost linear R-T relationship within a certain temperature range, T1 < T < T2. Such a linear function would take the form, Rearranging to bring temperature out gives, where a is the average temperature coefficient of resistance in the (T1,T2) temperature range, i.e., a is the slope of the R-T line. Both the measured temperature and the reference temparature should be within the (T1,T2) temperature range, T1 < {T, TRef} < T2
Glossary