|Strain Rosette for Strain Measurement|
A wire strain gage can effectively measure strain in only one direction. To determine the three independent components of plane strain, three linearly independent strain measures are needed, i.e., three strain gages positioned in a rosette-like layout.
Consider a strain rosette attached on the surface with an angle a from the x-axis. The rosette itself contains three strain gages with the internal angles b and g, as illustrated on the right.
Suppose that the strain measured from these three strain gages are ea, eb, and ec, respectively.
The following coordinate transformation equation is used to convert the longitudinal strain from each strain gage into strain expressed in the x-y coordinates,
Applying this equation to each of the three strain gages results in the following system of equations,
These equations are then used to solve for the three unknowns, ex, ey, and exy.
|Special Cases of Strain Rosette Layouts|